A syringe is a simple reciprocating pump consisting of a plunger (though in modern syringes, it is actually a piston) that fits tightly within a cylindrical tube called a barrel. The plunger can be linearly pulled and pushed along the inside of the tube, allowing the syringe to take in and expel liquid or gas through a discharge orifice at the front (open) end of the tube. The open end of the syringe may be fitted with a hypodermic needle, a nozzle or tubing to direct the flow into and out of the barrel. Syringes are frequently used in clinical medicine to administer injections, infuse intravenous therapy into the bloodstream, apply compounds such as glue or lubricant, and draw/measure liquids.
Sectors in the syringe and needle market include disposable and safety syringes, injection pens, needleless injectors, insulin pumps, and specialty needles. Hypodermic syringes are used with hypodermic needles to inject liquid or gases into body tissues, or to remove from the body. Injecting of air into a blood vessel is hazardous, as it may cause an air embolism; preventing embolisms by removing air from the syringe is one of the reasons for the familiar image of holding a hypodermic syringe pointing upward, tapping it, and expelling a small amount of liquid before an injection into the bloodstream.
The barrel of a syringe is made of plastic or glass, usually has graduated marks indicating the volume of fluid in the syringe, and is nearly always transparent. Glass syringes may be sterilized in an autoclave. However, most modern medical syringes are plastic with a rubber piston, because this type seals much better between the piston and the barrel and because they are cheap enough to dispose of after being used only once, reducing the risk of spreading blood-borne diseases. Reuse of needles and syringes has caused spread of diseases, especially HIV and hepatitis, among intravenous drug users. Syringes are also commonly reused by diabetics, as they can go through several in a day with multiple daily insulin injections, which becomes an affordability issue for many. Even though the syringe and needle are only used by a single person, this practice is still unsafe as it can introduce bacteria from the skin into the bloodstream and cause serious and sometimes lethal infections. In medical settings, single-use needles and syringes effectively reduce the risk of cross-contamination.
Medical syringes are sometimes used without a needle for orally administering liquid medicines to young children or animals, or milk to small young animals, because the dose can be measured accurately and it is easier to squirt the medicine into the subject's mouth instead of coaxing the subject to drink out of a measuring spoon.
Syringes come with a number of designs for the area in which the blade locks to the syringe body. Perhaps the most well known of these is the Luer lock, which simply twists the two together.
Bodies featuring a small, plain connection are known as slip tips and are useful for when the syringe is being connected to something not featuring a screw lock mechanism.
Similar to this is the catheter tip, which is essentially a slip tip but longer and tapered, making it good for pushing into things where there the plastic taper can form a tight seal. These can also be used for rinsing out wounds or large abscesses in veterinary use.
There is also an eccentric tip, where the nozzle at the end of the syringe is not in the centre of the syringe but at the side. This causes the blade attached to the syringe to lie almost in line with the walls of the syringe itself and they are used when the blade needs to get very close to parallel with the skin (when injecting into a surface vein or artery for example).
Standard U-100 insulin syringes
Syringes for insulin users are designed for standard U-100 insulin. The dilution of insulin is such that 1 mL of insulin fluid has 100 standard "units" of insulin. Since insulin vials are typically 10 mL, each vial has 1000 units.
Insulin syringes are made specifically for self injections and have friendly features:
- shorter needles, as insulin injections are subcutaneous (under the skin) rather than intramuscular,
- finer gauge needles, for less pain, and
- markings in insulin units to simplify drawing a measured dose of insulin.
- low dead space to reduce complications caused by improper drawing order of different insulin strengths.
Multishot needle syringes
There are needle syringes designed to reload from a built-in tank (container) after each injection, so they can make several or many injections on a filling. These are not used much in human medicine because of the risk of cross-infection via the needle. An exception is the personal insulin autoinjector used by diabetic patients.
Venom extraction syringes
Venom extraction syringes are different from standard syringes, because they usually do not puncture the wound. The most common types have a plastic nozzle which is placed over the affected area, and then the syringe piston is pulled back, creating a vacuum that allegedly sucks out the venom. Attempts to treat snakebites in this way are specifically advised against, as they are ineffective and can cause additional injury.
Syringes of this type are sometimes used for extracting human botfly larvae from the skin.
An oral syringe is a measuring instrument used to accurately measure doses of liquid medication, expressed in millilitres (mL). They do not have threaded tips, because no needle or other device needs to be screwed onto them. The contents are simply squirted or sucked from the syringe directly into the mouth of the person or animal.
The ancillary tool (generally part of a dental engine) used to supply water, compressed air or mist (formed by combination of water and compressed air) to the oral cavity for the purpose of irrigation (cleaning debris away from the area the dentist is working on), is also referred to as a dental syringe or a dental irrigation nozzle.
A 3-way syringe/nozzle has separate internal channels supplying air, water or a mist created by combining the pressurized air with the waterflow. The syringe tip can be separated from the main body and replaced when necessary.
In the UK and Ireland, manually operated hand syringes are used to inject lidocaine into patients' gums.
Governmental control of syringes
In some jurisdictions, the sale or possession of hypodermic syringes may be controlled or prohibited without a prescription, due to its potential use with illegal intravenous drugs.
The syringe has many non-medical applications.
Medical-grade disposable hypodermic syringes are often used in research laboratories for convenience and low cost. Another application is to use the needle tip to add liquids to very confined spaces, such as washing out some scientific apparatus. They are often used for measuring and transferring solvents and reagents where a high precision is not required. Alternatively, microliter syringes can be used to measure and dose chemicals very precisely by using a small diameter capillary as the syringe barrel.
The polyethylene construction of these disposable syringes usually makes them rather chemically resistant. There is, however, a risk of the contents of the syringes leaching plasticizers from the syringe material. Non-disposable glass syringes may be preferred where this is a problem. Glass syringes may also be preferred where a very high degree of precision is important (i.e. quantitative chemical analysis), because their engineering tolerances are lower and the plungers move more smoothly. In these applications, the transfer of pathogens is usually not an issue.
Used with a long needle or cannula, syringes are also useful for transferring fluids through rubber septa when atmospheric oxygen or moisture are being excluded. Examples include the transfer of air-sensitive or pyrophoric reagents such as phenylmagnesium bromide and n-butyllithium respectively. Glass syringes are also used to inject small samples for gas chromatography (1 μl) and mass spectrometry (10 μl). Syringe drivers may be used with the syringe as well.
Syringes may also be used when cooking meat to enhance flavor and texture by injecting juices inside the meat, and in baking to inject filling inside a pastry. It is common for these syringes to be made of stainless steel components, including the barrel. Such facilitates easy disassembly and cleaning.
Common workshop applications include injecting glue into tight spots to repair joints where disassembly is impractical or impossible; and injecting lubricants onto working surfaces without spilling.
Historically, large pumps that use reciprocating motion to pump water were referred to as syringes. Pumps of this type were used as early firefighting equipment.
There are fountain syringes where the liquid is in a bag or can and goes to the nozzle via a pipe. In earlier times, clyster syringes were used for that purpose.
Loose snus is often applied using modified syringes. The nozzle is removed so the opening is the width of the chamber. The snus can be packed tightly into the chamber and plunged into the upper lip. Syringes, called portioners, are also manufactured for this particular purpose.
- 9th century: The Iraqi/Egyptian surgeon Ammar ibn 'Ali al-Mawsili' created a syringe in the 9th century using a hollow glass tube, and suction to remove cataracts from patients' eyes, a practice that remained in use until at least the 13th century.
- Pre-Columbian Native Americans created early hypodermic needles and syringes using "hollow bird bones and small animal bladders".
- 1650: Blaise Pascal invented a syringe (not necessarily hypodermic) as an application of what is now called Pascal's law.
- 1844: Irish physician Francis Rynd invented the hollow needle and used it to make the first recorded subcutaneous injections, specifically a sedative to treat neuralgia.
- 1853: Charles Pravaz and Alexander Wood independently developed medical syringes with a needle fine enough to pierce the skin.
- 1865: Charles Hunter coined the term "hypodermic", developed an improvement to the syringe that locked the needle into place so that it would not be ejected from the end of the syringe when the plunger was depressed, and published research indicating that injections of pain relief could be given anywhere in the body, not just in the area of pain, and still be effective.
- 1867: The Medical and Chirurgical Society of London investigated whether injected narcotics had a general effect (as argued by Hunter) or whether they only worked locally (as argued by Wood). After conducting animal tests and soliciting opinions from the wider medical community, they firmly sided with Hunter.
- 1899: Letitia Mumford Geer of New York was granted a patent for "In a hand-syringe the combination of a cylinder, a piston and an operating-rod which is bent upon itself to form a smooth and rigid arm terminating in a handle, which, in its extreme positions, is located within reach of the fingers of the hand which holds the cylinder, thus permitting one hand to hold and operate the syringe..."
- 1946: Chance Brothers in Smethwick, West Midlands, England produced the first all-glass syringe with interchangeable barrel and plunger, thereby allowing mass-sterilisation of components without the need for matching them.
- 1949: Australian inventor Charles Rothauser created the world's first plastic, disposable hypodermic syringe at his Adelaide factory.
- 1951: Rothauser produced the first injection-moulded syringes made of polypropylene, a plastic that can be heat-sterilised. Millions were made for Australian and export markets.
- 1956: New Zealand pharmacist and inventor Colin Murdoch was granted New Zealand and Australian patents for a disposable plastic syringe.
- 1989: Manuel Jalón Corominas invented the auto-disposable needle-syringe. The head with the needle is detached from the syringe after use. This invention greatly reduced the spread of AIDS and other drug-use related diseases.
- Fire syringe has two meanings:
- Autoinjector, a device to ease injection, e.g. by the patient or other untrained personnel.
- Hippy Sippy
- Hypodermic needle
- Jet injector, injects without a needle, by squirting the injection fluid so fast that it makes a hole in the skin.
- Luer Taper, a standardized fitting system used for making leak-free connections between syringe tips and needles.
- Needle exchange programme, is a social policy based on the philosophy of harm reduction where injecting drug users (IDUs) can obtain hypodermic needles and associated injection equipment at little or no cost.
- Trypanophobia, a fairly common extreme fear of hypodermic syringes
- Syrette, similar to a syringe except that it has a closed flexible tube (like that used for toothpaste) instead of a rigid tube and piston.
- Syringing the ear to remove excess ear wax.
- Syrinx, the nymph from classical mythology after which syringes were supposedly named.
- Safety syringe, with features to prevent accidental needlesticks and reuse
- Vaginal syringe
- (in French) Swiss Telegraphic Agency, "Deux millions de personnes contaminées lors d'une injection à risque en 2014" Archived 2015-04-28 at the Wayback Machine, Radio télévision suisse, 23 February 2015 (page visited on 23 February 2015).
- http://www.thefreedictionary.com/dental+syringe "Disposable Needles & Syringes | Trigiene Dental | Quality Dental Supplies UK". Archived from the original on 2018-05-11. Retrieved 2017-08-25.
- "Disposable Syringe Markets". TriMark Publications, LLC. June 2013. Archived from the original on 2014-09-11.
- Bethany, Asked by. "Is it safe to reuse an insulin syringe? - CNN.com". Archived from the original on 2017-07-10.
- "Disposable Medical Supplies Markets". TriMark Publications, LLC. July 2014. Archived from the original on 2014-02-24.
- Sinding C (2010-12-08). "Making the unit of insulin: standards, clinical work, and industry, 1920–1925". Bull Hist Med. 76 (2): 231–70. doi:10.1353/bhm.2002.0097. PMID 12060790.
- "BD Diabetes-Insulin Syringe Needle Sizes". Bd.com. Archived from the original on 2011-09-28. Retrieved 2010-12-30.
- "BD Diabetes-Syringe Capacity and Dose Size". Bd.com. Archived from the original on 2011-07-07. Retrieved 2010-12-30.
- "Close-up of BD 1cc Syringes-UltraFine-30 Gauge-1/2", UltraFine II Short-31 Gauge-5/16" & MicroFine-28 Gauge-1/2"". Archived from the original on 2016-02-15. Retrieved 2010-12-30.
- "Diabetes Mellitus-Washington State University". Vetmed.wsu.edu. Archived from the original on 2011-01-03. Retrieved 2010-12-30.
- "ReliOn Insulin Syringe Markings". Relion.com. Archived from the original on 2010-11-28. Retrieved 2010-12-30.
- "Diabetes Health Syringe Listings" (PDF). Archived (PDF) from the original on 2011-07-10. Retrieved 2010-12-30.
- "Easy Touch Syringe demonstration-their syringes all measure in 1 unit increments". Easytouchsyringes.com. Archived from the original on 2011-07-10. Retrieved 2010-12-30.
- "Close-up of BD 1/2 cc Syringes—UltraFine-30 Gauge-1/2", UltraFine II Short-31 Gauge-5/16" & MicroFine-28 Gauge-1/2"". Archived from the original on 2016-02-15. Retrieved 2010-12-30.
- "Ulti-Care U100 Syringes-Product Information". Ulti-care.com. Archived from the original on 2011-07-17. Retrieved 2010-12-30.
- "Easy Touch Syringes demonstration-all syringes with 1 unit increment markings". Easytouchsyringes.com. Archived from the original on 2011-07-10. Retrieved 2010-12-30.
- "Close-up of BD 3/10 Syringes-UltraFine-30 Gauge-1/2", UltraFine II Short-31 Gauge-5/16", UltraFine Short-Half Unit Markings-31 Gauge-5/16" & MicroFine-28 Gauge-1/2"". Archived from the original on 2016-02-15. Retrieved 2010-12-30.
- "Easy Touch Syringes-all products measure in 1 unit increment markings". Easytouchsyringes.com. Archived from the original on 2011-07-10. Retrieved 2010-12-30.
- "Comparison of 3/10 cc syringe marks—half unit and whole unit scale-BD". Archived from the original on 2011-09-28. Retrieved 2010-12-30.
- Bush MD, Sean P. (February 2004). "Snakebite Suction Devices Don't Remove Venom: They Just Suck" (PDF). Annals of Emergency Medicine. 43 (2): 187–188. doi:10.1016/j.annemergmed.2003.10.031. PMID 14747806. Archived (PDF) from the original on 2013-05-17. Retrieved 2014-12-24.
- Boggild, Andrea K.; Jay S. Keystone; Kevin C. Kain (August 2002). "Furuncular myiasis: a simple and rapid method for extraction of intact Dermatobia hominis larvae". Clinical Infectious Diseases. 35 (3): 336–338. doi:10.1086/341493. PMID 12115102. Retrieved 2014-09-28.
- "How to use your oral syringe". Netdoctor. 2012-11-23. Archived from the original on 2011-01-04.
- "Lidocaine Hydrochloride (Local) Monograph for Professionals - Drugs.com". Archived from the original on 2017-07-29.
- Zakrzewska, J. M.; Boon, E. C. (23 August 2003). "Use of safety dental syringes in British and Irish dental schools". British Dental Journal. 195 (4): 207–209. doi:10.1038/sj.bdj.4810445. PMID 12970703.
- Zakrzewska, J. M.; Greenwood, I.; Jackson, J. (27 January 2001). "Cross-infection control: Introducing safety syringes into a UK dental school – a controlled study". British Dental Journal. 190 (2): 88–92. doi:10.1038/sj.bdj.4800891. PMID 11213339.
- "Children With Diabetes-Prescription Needed or Not?". Childrenwithdiabetes.com. Archived from the original on 2010-12-14. Retrieved 2010-12-30.
- "Criminal Law and Syringe Disposal in Delaware". Archived from the original on 2013-03-30. Retrieved 2012-04-03.
- Finger, Stanley (1994). Origins of Neuroscience: A History of Explorations Into Brain Function. Oxford University Press. p. 70. ISBN 978-0-19-514694-3.
- "16 Indian Innovations: From Popcorn to Parkas". news.nationalgeographic.com. Archived from the original on 2017-10-16. Retrieved 2017-12-25.
- "10 Native Inventions and Innovations That Changed the World - Indian Country Media Network". indiancountrymedianetwork.com. Archived from the original on 2017-12-25. Retrieved 2017-12-25.
- Brunton, D. (2000). "A Question of Priority: Alexander Wood, Charles Hunter and the Hypodermic Method". Proceedings of the Royal College of Physicians of Edinburgh. 30: 349–351.
- "Patent US622848 – Letitia Mumford Geer". Archived from the original on 2015-02-22.
- Walford, Cornelius. The insurance cyclopaedia: being a dictionary of the definitions of terms used in connexion with the theory and practice of insurance in all its branches: a biographical summary ... a bibliographical repertory of all works written upon the subject ... an . London: C. and E. Layton ;, 1871,1880. 337 Print. Walford, Cornelius (1874). "The Insurance Cyclopeadia: Being a Dictionary of the Definitions of Terms Used in Connexion with the Theory and Practice of Insurance in All its Branches: A Biographical Summary ... A Bibliographical Repertory of All Works Written Upon the Subject ... An Historical Treasury of Events and Circumstances Connected with the Origin and Progress of Insurance". Archived from the original on 2017-02-23. Retrieved 2016-10-29.
- Hans-Jürgen Bässler und Frank Lehmann : Containment Technology: Progress in the Pharmaceutical and Food Processing Industry. Springer, Berlin 2013, ISBN 978-3642392917
|Wikimedia Commons has media related to Syringe.|
|Look up syringe in Wiktionary, the free dictionary.|